The Synergy HT Multi-detection reader is a robotic-compatible microplate .. and J. Sambrook () Molecular Cloning A Laboratory Manual, Cold Spring. View and Download BioTek Synergy HT operator’s manual online. Multi-Mode Microplate Reader. Synergy HT Laboratory Equipment pdf manual download. Manuals and User Guides for BioTek Synergy HT. We have 2 BioTek Synergy HT manuals available for free PDF download: Operator’s Manual.
|Published (Last):||1 January 2014|
|PDF File Size:||2.45 Mb|
|ePub File Size:||15.6 Mb|
|Price:||Free* [*Free Regsitration Required]|
Saturation Errors If you suspect that problems occurred during shipment, if you received the reader back Place the tip priming trough in the left-rear pocket of the carrier. Wipe with lens-cleaning tissue and set aside.
BIOTEK SYNERGY HT OPERATOR’S MANUAL Pdf Download.
Page 97 H12 position is 1. This error indicates that during a self-test, the lamp did not indicate that it was off. The trough holds about 1. Page Chapter 5: Sybergy both of them and set them aside.
Liquid Test 2 Install Gen5 Software Save all packaging materials. Chapter 1 Introduction This chapter introduces the Manuak HT, describes its key features, lists its package contents, and provides contact information for technical assistance.
Shake time is programmable by the user. Examine the fittings; no leaks should be detected. Models With Injectors Click to close the Data Reduction dialog. See criteria in text below. By identifying the wells by their dispense volumes in the Plate Layout, Gen5 will calculate these values syneegy you.
The calibration sticker is affixed directly to the plate. Other Errors 2a Inspect the block for spills or other contamination.
Remove two screws from the top center of the rear panel Remove two side screws When reinstalling the shroud, press down firmly on the top to maintain a good light ,anual while tightening the top screws. Otherwise, the results of the read may be invalid. Troubleshooting Troubleshooting If any tests fail, please try the suggestions below. Slide the filter wheel back into its chamber.
Introduction Package Contents Part numbers and package contents are subject to change. Depress tab to separate the connectors Biootek the thumbscrew located in the left rear of the instrument and set it aside.
BioTek Synergy HT Manuals
Technical Assistance Center tac Usa Fcc Class A Page 55 Insert the accessories box alongside the dispense module box. This exposes the ground wire. Remove The Shipping Panel See the photos on the next page.
Unpack and Inspect the Reader Page 34 – 2: Use two people when lifting and carrying the instrument. See Figure 3 on biotk next page. The TR cartridge allows light from the xenon flash bulb to be input to the fluorescence optical system within the Synergy instrument.
Alternate Test Solutions Put the tubes back in. Stock Solution A or B, which may be formulated by diluting a dye solution available from BioTek A manjal from the ingredients listed below B. Contact Information outside the U. Precautions Precautions The following precautions are provided to help avoid damage to the instrument: Page 30 seconds, and then turning the instrument back on. Microplates should be perfectly clean and free of dust or bottom scratches.
Canadian Department of Communications Class A